tikabass: Nice and useful demonstration, it's great you take the time to do this. A
couple of notes: in actual practice, you really need to zoom in more on the
front of the waveform to get a more precise measurement, Also, it is often
more practical (and faster to do) to increase/decrease the frequency of the
wave generator to match the resistor / capacitance pair, intead of changing
the resistor value (which could be a specially selected resistor).
Jim C: Enjoy your tutorials!! Keep going!! Great approach and delivery.
Additionally, any advice on purchasing my first oscilloscope? I do not have
a lot of funds for this but I do want to purchase something that will be
useful over the long run. Any advice is greatly appreciated! Thank you!
alex castro: ...physical resistance adds to the impedance of the generator 50 ohms? or I
can remove the physical resistance and do it with the output impedance of
the function generator 50 ohms?... and your function generator is set to
square wave with offset positive DC to reference to ground? 0v to 1v for
alex castro: and other question, your function generator, output impedance is 50 ohms ?,
and your oscilloscope is to 1 Mohm ?? this difference is correct? and not
Affect end to measure ??.. thanks for your videos.
Bill Stewart: Sir- Can you give me advice on buying an Oscilloscopes? I can't justify a
large amount of money for this until I become more efficeint circuit
building, design and testing. Looking for entry level to electronics that
would be a good purchase to further my knowledge of the finer points of
Oscilloscope. Thanks in advance! Bill
AlainHubert: To marbleshark: FYI Martin is making a living out of his YouTube videos
So he's not donating anything. He's sharing knowledge and information in
exchange for money.
Daemonforce Gaming: Awsum video very understandable and educationall and im trying to learn
basic electronics love the way you explain things keep up the good work
Steive Comfort: Exxcellent, keep up the good work.
TheThore: So you need a calculator for dividing by 1000? ;)
Andre Sant'Anna: Great video.
Like many people have said, when you are measuring such a tiny tiny value
capacitor, then there are a ton of other stray capacitances that confound
the measurement - the probe, the wiring, the breadboard, they all have
values in the pF range so you are actually trying to do something pretty
difficult. These stray values don't affect the measurement of the bigger
capacitors much because the stray values are so much smaller in comparison.
When measuring something small like this, you might want to connect a bunch
of them in parallel (the capacitances will add) and then divide by the
number of capacitors you used to get the answer for a single one of them.
Roman Kyselý: Martin, thx. All videos was for me realy usefull. Keep going...
Flapjackbatter: "How to measure a capacitor with an oscilloscope."
Look at the time. 46 minutes. You babble way too much about things that are
321reh: Very Well Done Video..Thanks Martin!!!!!!!!!!!!!!
Samuel Esther: Oh my god, 46mins just to explain how to measure a capacitor value with an
oscilloscope!.. it could be explained in 4 times less, which would be 4
times more interesting. You've made a simple and passionate subject heavy
and complicated to understand. Keep it simple and short please.
Curvian Vynes: Hey, 5.99E^-11 is not 5.99^-11. It's 5.99*10^-11.
Anyway, great video! Thanks!
marbleshark: I love this guy - thanks so much Martin for generously donating your time
to educating others with your excellent tutorial videos... I've learnt a
lot and it's set me well on the way :) Cheers
magicman33069: Consider if you will, the capacitance of the probe, as well as the loading
effect of test equipment used, which will of course, effect your results.
shabmw tata: use a pico tool
Najim MOUADILI: How to mesure permittivity of materials using osiloscope
Mike O'Neill: Could the error be caused by the noise induced? Noise is just another wave
form placed over the one you are looking at. So, you should be taking your
voltage readings either from the top of the noise, or the bottom. Not the
top and bottom of the entire wave form.
GardeningZ: Nice measurement.
Now can we use a current mode square wave generator to measure
capacitance? Say if I have a programmable power supply?
Ahmed Abdalsalam: thanks for your effort.
Jeff Burris: I have o-scopes but I don't have a dedicated function generator per say (as
a test instrument made for the purpose). Until I do (like tonight, now, in
a pinch ) perhaps this video can get me testing capacitance nonetheless,
using something on hand, like maybe a receiver or transmitter receiving or
transmitting a plain carrier or consistent signal (not Beethoven's 5th) :)
I'm not trying to calibrate instruments, rather, I just need to get
approximate readings on hand-made air capacitors for value. I want to make
a rotary but instead of flat vanes use concave soda can bottoms (novelty,
for antenna tuning with a coil)... can bottoms for both stator & rotor
vanes (I know I can't offset them but spindle through the middle should be
fine :) :) o man I can maybe measure inductance of the coil, too, without
buying an LCR meter this week (my multi has neither) THANK YOU
(subscribing). Between you and Ray Heffer I'm in high cotton.
John Elliott: Thanks for trick and hint and maths : 0 )
Dustin Lewis: I've tried to watch a few of your videos, but you talk so much, its
difficult to string the important bits together between the babbling.
The Lightning Stalker: Stray inductances are going to affect your measurement, particularly with
small value capactiors. This should be taken into account. Oscilloscope
probes also have a capacitance which will also affect the reading. Your
breadboard also has a capacitance.
a11457exp: awesome ..
Totardo Tobing: Too slow :(
tim master: How large of capacitors could you use this technique with?
Phil Brown: You explained that very well ... Subscribed
charles bell: Just Great !!
Round House: Reducing the lead on the resister will also help. Nice video.
tim master: I wonder if a cage would help reduce the noise.
Rod Dormido: the possible source of error is that you forgot to factor in the
capacitance of the oscilloscope probe.
Barry Moore: An alternative measurement method is to use a sine wave and a two channel
scope, Place the channel 2 across the cap, and the channel 1 across the
resistor (R) and capacitor (C) in series. Adjust the frequency (F)
until channel 2 is 0.707 of the input (channel 1 value), At this frequency
the capacitive reactance is equal to the resistance of the resistor i.e.
R=1/(2*pi*F*C). As we know R and F we can calculate C. A digital storage
oscilloscope will display accurate peak voltages of the channels. The phase
angle could also be measured as it is exactly 45degrees.
Gabriele Barbaraci: finally I have seen someone explaining the things how must be done!!!!
Chakir El yattafti: Thank you very much. I do believe that the problem you have with 4.7 pF is
due to the reactance of the capacitor become significant big due to this
formula reactance is Xc=1/W.C the reason you did not have a problem with
47nF is because the reactance was low. your Circuit Impedance is
Z=sqrt(R^2+1/W^2*C^2), in order to compensate for the reactance of the
capacitor try to increase the frequency until the term 1/W^2*C^2 in Z
equation get smaller in order to get the impedance as pure resistive as
possible. I hope that helps.You do really great job.
Nouman tajik: Thanks for tutorials, It is Really Helpful, Good Luck :)
Paul Collins: I happy to say I enjoyed the video very much
Anthony Stewart: Dear MjL,
The probe capacitance must be observed when measuring capacitance <
100pF. Probe settings for 10:1 will give a higher impedance than 1:1 and
thus lower capacitance.
Harish Kumar: nice one! thank u
Chris Gonzales: This was a great exercise. Even with a very basic scope I got reasonable
results ( with practice ). If I had to identify a bag of small "mystery"
caps I would put a bunch in parallel to reduce the error.
Bartolomeo Cianciatella: So when the square wave is down to 0, the capacitor is discharging itself
through the resistor and INSIDE the function generator ? Is that ok ? What
would be happen with higher voltages/currents ?
I suppose you could put a diode in series, and take its voltage drop into
Just asking because I'm pretty new to these things.
supershwa: An impressive arsenal of tools on your workbench, and lots of know-how! I
think a 5 minute version based on the [title] of the video would be
helpful, but the long version does show you're qualified as a
Anonymous News: what is all the noise on the scope
Marcos Ramirez: thank you so much, electronics is amazing!!
zinou sid: thank you so much (good job) (y)
logicsystems paddy: GREAT . VERY NICE WAY EXPLAINED
Aaron Hayes: The capacitance of the cables leads and breadboard may well higher than
that 4.7pF capacitor.