C & S Electronics: what is the impedance of Oscilloscope Probe coaxial is it 50 ohms ?
Lenient: I've only just found this video and this is the best probe tutorial by far.
DesignMicros: Great explanation, clear and to the point. I'm recommending this to our
entry level techs!
I've also seen some of your other videos and am impressed with your
hweenmask: uhhh ... so... wtf is 1X and 10X ?
8233Eire: The worst ever....unless you have a PhD!!!
InnaSoulSounds: This guy is the bee's knees, really.
Mike Novo: Best tutorial on this ever!
Fusion Music: Great video. There are a lot of very clever people on you tube but can be a
bit laboured in their delivery. This is well thought out and said with
confidence. It was a pleasure to listen to and subscribe.
C & S Electronics: nicely explained
asuspower: Wow, this is amazing, I understand this now!! :) :) thank you so much :)
Jared Kusner: Had fun adjusting several probes to my scope. Thanks for making it easy to
Tesla Explorer: Great video! I finally got an old tek, never used a scope before. Was
wondering what the 5v pwm "ring" was. Now I do! We sometimes forget that we
live in an imperfect world!
Axel Haar: Very good to know. I compensated my probes to my oscilloscope.
Excellent. Thanks a lot!
Washingstone: Great thank you.
I like quality channels like this one.
Thumbed up and subscribed.
sean craig: I've always wondered why the leading edge of the square wave changes but
not the trailing edge, it stay nice and sharp?
superdau: How does the trimmeable bypass cap work when it's near the socket, but the
1x/10x selector switch is in the probe? How can they be in parallel in the
way you've drawn them?
Tony T.: Great lesson on a key subject! Thank You!
GulfSouthGene: Using Oscilloscopes most of my engineering career, been to Tektronix
schools, etc. but your explanations, easy to understand and intuitive. Good
Job 73's KC5WS
Samuel Atkins: I need a 100X probe for dso520 oscilloscope. Can i use any 100X probe?
Mustafa Tahsin Güler: Hello, I have uploded a video about my question. If you watch and tell me
what is happening, I would be very grateful.
Grounding type and probe difference having a signal
Abrar Shaikh: One of the most amazing thing on YouTube for engineers and hobbyist. I have
planned to watch all this videos once my semester exams are over. You are
amazing. I never got to see output in our lab scope, and even when I got to
see, it was accidental and didn't know how to reproduce it. I'm pretty sure
this tutorials of yours will train me in becoming a better engineer.
Хлебников Денис: I have to do my own divider, thank you a lot.... You explained everything
and showed i needed!
Mustafa Tahsin Güler: Does stray capacitance come from the interaction between core wire and
grounded wire in the coaxial cable ? I made 2 probe cable, one from a rg58
and the other from a rg174, rg174 which is thiner has higher noise, what
may be the reason ? But I think, they are noy because of capacitive loading
if its effect is as shown here at the end of the video, here I am talking
about parasitic noise distorting the signal.
Chris Greene: So, I was trying to measure a weak signal (about 2 mV [Max resolution on my
scope is about .2 mV)) in a fairly simple circuit. Couldn't figure out why
I wasn't seeing anything. Then I realized my probe was on 10x.
Basic 1X and 10X Oscilloscope Probe tutorial
Clifton Jamison: Great tutorial,I learned a lot ,thanks.My uncle's call letters were
W2UEN,and he was in Plattsburgh,N.Y.He would repair everyone's TV's in the
area,when they first came out.He had one of the first TV's that were
available.He was a character.
Jolita J: Thank you very much for this video. Helped unbelievably.
w2aew: @Afrotechmods Thank you! I'm glad you liked it!
Miguel Rodriguez: Thanks. The main cables are 2 meters long, for normal use I think it is
excesive, does tektronix make pasive probes with shorter main cables?
Chris Hinton: you know what -- it's the simple things that i missed in school -- i've
never thought about calibrating an o'scope (calibrating a VNA, yes --- but
an o'scope never crossed my mind). anyways, thanks for the video! very
w2aew: @realy1985 Thank you very much!!
bluebelt4life: what should the value of the trimmer cap be? For example with 20pF scope
w2aew: In an ideal world, for a 10X probe, the compensation capacitance would be
~2pF. In reality, it will be different because you have to account for the
cable's capacitance and other parasitics.
zloben9000: i found this site dealextreme com they have a 60MHz Oscilloscope Probe with
X1/X10 Switch (Max 600V) for 6 dollars with free shipping .. i just ordered
kc8rwr: I knew my probes had an adjustment but I left it alone because I didn't
know how/when to adjust them. Now when I get home I will be checking mine.
Max Manner: Brilliant teacher!
foundede: Is there an "awkward" stage between when the signal sees 10X attenuation by
the 10 megaohms of pure resistance and when the signal frequency increases
high enough to see the 10X attenuation by the input and compensation
HammerToneAmps: Excellent basic tutorial .. very helpful. Thanks
w2aew: There are 10x, 20x, 100x, etc. probes designed for a 50 ohm impedance
input, but they are hard to come by. However, I find it odd that your scope
would have ONLY a 50 ohm input, especially for a scope that has only 100MHz
BW. I would expect that it can be switched to have a 1Mohm input as well,
in which case you can use ordinary 10x passive probes.
w2aew: @sdscotto If I had access to some active probes and current probes, I
would. There are some videos and webinars online. Go to tek.com / power
(without the spaces) then select Library and Videos & Webinars. I'd like to
do one of my amateur videos on this, but I just don't have the equipment.
Erwin Ried: Hi! My scope says 400V max input, can I connect the probes to Mains? 220V
using 10X? or I will blow the probes?
321reh: Keep churning those Videos Out,,,I love to learn!!
BombshellBill: great video man! you really know you're stuff!
65Superhawk: Yep. It was there. I didn't see it at first. Most probes I saw on Youtube
had a differently colored "box" near the BNC connector. This one was not a
box, but merely a part of the molded black plastic sleeve that was bigger
and had a screw inset within it. Thanks!
w2aew: @HCI344 I'm glad you found it useful!
321reh: Thanks for all the Great videos you put out!!! You should do a Video on
Consumer Electronics Repair. MR.OHM enjoys ALL your Videos,,,Thanks!!
Afrotechmods: Most comprehensive video on probes I have ever seen. Hats off to you sir!
w2aew: @ElectronixCore Thank you very much! Please let me know if there are other
similar topics that you'd like to see future videos on!
nikola_tesla: @w2aew Will do! Thanks so much!
Nemanja Todorovic: This is what I thought. Superb video! Covered is everything you need for a
beginner. Thank you!!!